P6100 100 MHz Oscilloscope 1x & 10x Passive Probebrand: SainSmart SKU: 20-010-204
Attenuation Ratio: 1x & 10x
Input Resistance(MΩ): 1x: 1
Input Capacitance(pF): 1x: 90
Compensation Range(pF): 10x: 10-35
Brandwidth(MHz): 1x: 6
Rise time(ns): 1x: 58
Working Voltage(Vp-p): 1x: 300
Net Weight(g): 64
Humidity: â‰?5% Relative Humidity
These high passive impedance probes are designed and calibrated for use on any oscilloscope that has an input impedance of 1 MΩ and whose input capacity is within the compensation range (refer to the specifications.) When connecting the probe, please connect it to the oscilloscope before testing signals. When disconnecting the probe, first disconnect the probe tip form the test signal. In the process of test, make sure that alligator clip can be grounded reliably.
Low-frequency Compensation Adjustment
Low frequency response can be matched to the oscilloscope by adjusting the compesation trimmer on the head of the probe.
1. Connect the probe to the oscilloscope and to a 1KHz square waveform source.
2. Let the oscilloscope display a stable waveform.
3. Carefullly adjust the trimmer tool to botain the flat test tops to the square waves displayed on the oscilloscope